I have the following message in Omnilogic:
CHLOR Too Low Salt Chlorinator1
I guess the technical term is that it is an ALARM in Omnilogic.
Any idea what that is referring to? What is the remedy? If I look at the Omnilogic control panel via online internet access (not through mobile app), there in an indication of "Chlorination paused - low salt".
What do i need to do to get the salt higher and the chlorination going again?
Are answers to these questions in ANY Hayward literature? If yes, please point me in the right direction.
Instant salt 1593
Average salt 2185
Relay polarity K2
Cell voltage 28.13
Cell Current 4.33
Cell Type T15