I am going on year to of my IGP w/ SWG. Since starting the system up several weeks ago, I have been getting low salt readings on my PL-PS-8. It was reading between 2300 and 2800 PPM. So I took the T-Cell-15 off and cleaned with 4 parts water, 1 part muratic acid for 10 minutes. Rinsed and reinstalled. Initially I got my reading up to 3000, which is very close to my 3100 PPM I get with my Taylor salt test kit. But after 24 hours, it is once again reading 2300PPM. Water temp is 68F.
Diags read as follows: 27.21v -4.80A 2300PPM 65F
Any other suggestions? I realize that I could add more salt to get actual up to 3500 and probably stop the check system errors, but the reading is still going to be way off Taylor actual.
Thanks for your help!